Информация по оборудованию:
|Solar Manufacturing Corporation QCA Analyzer|
Solar Manufacturing Corporation
|Описание:||Quick Check Capacitor Analyzer|
The Solar QCA Capacitor Analyzer provides radio service engineers and laboratories with a capacitor Analyzer that combines in one instrument a convenient and simple method of making a complete analysis of condensers under conditions of operation, static or dynamic. The performance characteristics of radio and video receivers have imposed requirements which can be maintained only by the use of high quality component parts. These parts, especially condensers, must have the proper electrical characteristics to perform their designated functions properly, if satisfactory results are to be obtained from receivers. Condensers have a detrimental effect on reception long before they become entirely defective. The problem of determining the quality or operating efficiency of a condenser requires a complete analysis of its constants. In this instrument condensers may be tested for capacity, leakage, insulation resistance, power factor, opens, shorts, intermittents, and r.f. impedance. Also with the full development of the Quick-Check features plus the Wien Bridge, we find it possible to present an instrument which will give every qualitative and quantitative measurement of capacitors necessary in modern servicing of a simple indication. model QCA tells the performance story of all types of paper, mica, electrolytic, trimmer and air condensers. Checking for quality is done under conditions of either static or dynamic performance; the condenser under test may be operating in a circuit, or it may be disconnected. Accurately calibrated markings on the panel are provided for convenient measurements of capacity. These tests are made on a Wien Bridge with three ranges by multiplying the direct scale reading by the bridge constant. The power factor of all types of electrolytic capacitors is indicated by the degree of opening of the electron-eye. The capacity range of the instrument is from .00002 mfd. to 70 mfd. Only two terminals are used for all tests. The switch in the QCA Analyzer automatically connects the unit under test into the proper circuit. The leakage test part of the instrument consists of a D.C. Power supply and electron-ray tube. A selector switch is provided with voltage settings covering practically all electrolytic condensers which may be encountered in the field. The insulation resistance range of the instrument provides quick and positive means of determining faulty condensers. The quick-check feature of the QCA consists essentially of a complex balanced radio frequency oscillator, an electron-ray tube indicator and a source of power.
Эти руководства доступны для вышеуказанного оборудования:
|Solar Manufacturing Corporation -- QCA -- Обслуживание и Руководство пользователя|
|Руководство Тип:||Обслуживание и Руководство пользователя|
|Размер:||10.94 Mbytes (11468377 Bytes)|
|Качество:||Сканированный документ, читая частично плохо, частично не читается.|
|Дата публикации:||15 Ноябрь 2015|
|Скачиваний:||297 начиная с 15 Ноябрь 2015|
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.