Информация по оборудованию:
|Krohn-Hite Corporation 3200 R Другой|
|Описание:||SOLID STATE VARIABLE FILTER|
The Models 3200 and 3202, illustrated in Figure 1, are solid state variable electronic filters with cutoff frequencies continuously adjustable over the frequency range from 20 Hz to 2 MHz. The pass-band gain is unity (0 db), with an attenuation rate of 24 db per octave outside the pass-band. Maximum attenuation is greater than 80 db and the output hum and noise is less than 100 microvolts. The Model 3200 can function as either a High-Pass or Low-Pass Filter. In the High-Pass mode of operation the maximum input signal is 3 volts rms and the upper 3 db point occurs >t approximately 10 MHz. In the Low-Pass mode the Filter is direct-coupled and the combined ac plus dc input signal should not exceed 4,2 volts peak. The Model 3202 consists of two Model 3200’s mounted in a single cabinet isolated from each other with independent power supplies, and input and output connectors. When these wo filter channels are switched to the same mode of operation and connected in series with both dials set to the same cutoff frequency, the Model 3202 will function as a High-Pass or Low^Pass Filter with an attenuation rate of 48 db per octave. If the two channels are connected in series, and one channel is operated in the Low-Pass mode and the other channel a the High-Pass mode, the Model 3202 will function as a Band-Pass Filter with attenuation ' rate of 24 db per octave outside the pass-band. When the two channels are connected in parallel, as described in Section 2. 3, the Model ^202 will function as either a Band-Reject Filter with cutoff frequency limits from 20 Hz to 2 MHz or provide a Sharp null at any frequency between 40 Hz and 800 kHz. This Filter has a maximally flat or Butterworth characteristic when the RESPONSE switch(s), located on the rear of the chassis, is in the MAX FLAT position. For pulse-type waveforms this switch should be in the SIMPLE RC position, optimum for transent-free filtering.
Эти руководства доступны для вышеуказанного оборудования:
|Krohn-Hite Corporation -- 3200 R -- Обслуживание и Руководство пользователя|
|Руководство Тип:||Обслуживание и Руководство пользователя|
|Размер:||1.43 Mbytes (1500197 Bytes)|
|Качество:||Сканированный документ, читая частично плохо, частично не читается.|
|Дата публикации:||06 Сентябрь 2015|
|Скачиваний:||237 начиная с 06 Сентябрь 2015|
OPERATING AND MAINTENANCE MANUAL GENERAL DESCRIPTION ... 1 OPERATION ... . ... 7 CIRCUIT DESCRIPTION ... 10 MAINTENANCE . . . . . . ... . . 13 CALIBRATION AND ADJUSTMENT ...
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.