Tektronix TDS 600C Osciloscópio
Nenhuma foto disponível!
Talvez você possa carregar uma foto para os
Tektronix TDS 600C ?
Se você tem outros manuais para os
Tektronix TDS 600C você pode carregar os arquivos aqui .
The TDS 500D
and TDS 700D
portable, four-channel instruments suitable for use in a
variety of test and measurement applications and systems.
Use a combination of front-panel buttons, knobs, and
on-screen menus to control
the many functions of the Oscilloscope
. The front-panel
controls are grouped
according to function: vertical, horizontal, trigger, and
special. Set a function you
adjust often, such as vertical positioning or the time base
setting, directly by its
own front-panel knob. Set a function you change less often,
such as vertical
coupling or horizontal mode, indirectly using a selected menu.
Pressing one (sometimes two) front-panel button(s), such as
displays a main menu of related functions, such as coupling
and bandwidth, at
the bottom of the screen. Pressing a main-menu button, such
displays a side menu of settings for that function, such as
AC, DC, or GND
(ground) coupling, at the right side of the screen. Pressing
a side-menu button
selects a setting such as DC.
On-screen readouts help you keep track of the settings for
such as vertical and horizontal scale and trigger level.
Some readouts use the
cursors or the automatic parameter extraction feature
(called measure) to display
the results of measurements made or the status of the
General Purpose Knob
Assign the general purpose knob to adjust a selected
parameter function. More
quickly change parameters by toggling the SHIFT button. Use
the same method
as for selecting a function, except the final side-menu
selection assigns the
general purpose knob to adjust some function, such as the
position of measure-
ment cursors on screen, or the setting for a channel fine gain.
The user Interface
also makes use of a GUI, or Graphical
, to make
setting functions and interpreting the display more
intuitive. Some menus and
status are displayed using iconic representations of
function settings, such as
those shown here for full, 250 MHz and 20 MHz bandwidth.
Such icons allow
you to more readily determine status or the available settings.
Signal Acquisition System
The signal acquisition system provides up to four,
full-featured vertical channels
with calibrated vertical scale factors from 1 mV to 10 V per
on TDS model. All channels can be acquired simultaneously.
Each of the full-featured channels can be displayed,
vertically positioned, and
offset, and their vertical coupling specified. Some models
can have their
bandwidth limited (250 MHz or 20 MHz). Fine gain can also be
Besides these channels, up to three math waveforms and four
forms are available for display. (A math waveform results
when you specify dual
waveform operations, such as add, on any two channels. A
results when you save a waveform in a reference memory).
There are three horizontal display modes: main only, main
delayed only. You can select among various horizontal record
A feature called “Fit to Screen” allows you to view entire
within the 10 division screen area. Waveforms are compressed
to fit on the
screen. See Table 1–2.
Both the delayed only display and the intensified zone on
the main intensified
display may be delayed by time with respect to the main
trigger. Both can be set
to display immediately after the delay (delayed runs after
main mode). The
delayed display can also be set to display at the first
valid trigger after the delay
The delayed display (or the intensified zone) may also be
delayed by a selected
number of events. In this case, the events source is the
The delayed trigger can also be set to occur after a number
of events plus an
amount of time.
Tipo de manual:
Manual de serviço
4.28 Mbytes (4491816 Bytes)
ID de manual:
Documento eletrônico, digitalização, bem legível.
Data de upload:
25 Dezembro 2017
This manual is divided into sections, such as Specifications
and Theory of
Operation. Further, it is divided into subsections, such as
and Removal and Installation Procedures.
Sections containing procedures also contain introductions to
those procedures. Be
sure to read these introductions because they provide
information needed to do
the service correctly and efficiently. The following is a
brief description of each
Specifications contains a product description of the TDS
500D, TDS 600C
and TDS 714L
Oscilloscopes and tables of the
descriptions that apply to it.
Operating Information includes general information and
tions at the level needed to safely power on and service
statement of the service strategy that this manual supports,
for shipment of the oscilloscopes are found in this section.
Theory of Operation contains circuit descriptions that
support general service
and fault isolation.
Performance Verification contains a collection of
procedures for confirming
that these oscilloscopes function properly and meet
Adjustment Procedures contains a collection of procedures
these oscilloscopes to meet warranted limits.
Maintenance contains information and procedures for doing
corrective maintenance of these oscilloscopes. Instructions
for cleaning, for
module removal and installation, and for fault isolation to
a module are
Options contains information on servicing any of the
options that may be present in your Oscilloscope
Electrical Parts List contains a statement referring you to
Replaceable Parts, where both electrical and mechanical
modules are listed.
Diagrams contains a block diagram and an interconnection
for isolating failed modules.
Mechanical Parts List includes a table of all replaceable
descriptions, and their Tektronix