Skanowany dokument, wszystko czytelne.
The present Service Manual applies to both the 9400
the 9400À DSOs. Both models are called 9400
information applies equally to both oscilloscopes.
Where the two models have distinct features, the model
is explicitly stated.
The chief purpose of this manual is to provide information
for technicians, mainly authorized LeCroy
personnel, who are responsible for repairs and modifications
to the LeCroy 9400
To this end, the descriptions provided are intended to be
of sufficient depth to enable faults to be diagnosed to
the level of the relevant board, so that the customer can
be quickly supported by exchange of boards. In many cases
the fault may be corrected at the local LeCroy
but there are several areas of the circuitry where
replacement of a part would need to be followed by
calibration which could be done only with specialized
equipment available at the main LeCroy
Calibration procedures are given in this manual only
for those areas which could be serviced locally.
A DSO repair intervention at board exchange level
can only be done by qualified technicians who have
followed the basic 9400
.9400A service training.
also offers these courses to customers. In
addition, there is a very comprehensive 9400
and Calibration Software
Package CALS0FT (order codes
CS01/CS02) based on the IBM PC or a compatible computer.
package is also available to customers,
as well as all the hardware making up the 9400
Calibration System (order code CS-S) which is used
in all LeCroy
service offices to ensure full performance
of the instrument. This system provides Calibration
Certificates traceable to NBS. LeCroy
also offers training
classes on DSO calibration.
This manual could be improved by the inclusion of useful
information resulting from detection and correction
of faults in 9400
.9400A DSOs at any LeCroy
Each time a DSO is opened, an official LeCroy
report should be sent to LeCroy
Customer Service. The information is entered into
a centrally maintained DSO data base for failure
analysis and engineering feedback.
Before undertaking any work on the 9400
DSO you should
read the next sections - WARNINGS and VALID RANGE
of FIELD MAINTENANCE.