Skanowany dokument, wszystko czytelne.
Including Options 004 and 005
This manual applies directly to HP Model 86290B RF Plug-In
having serial number prefix 2227A.
With changes described in Section VII, this manual also applies to instruments with serial numbers prefixed 1704A, 1727A, 1737A, 1742A
1807A, 1840A, 1847A, 1852A, 1904A, 1908A, 1933A, 1952A, 2021A, 2034A, 2046A, 2109A, 2138A, and 2217A.
This Operating and Service manual contains information required to install, operate, test, adjust, and service the Hewlett-Packard Model 86290B RF PlugIn. Figure 1-1 shows the instrument and Accessories
supplied. This section covers instrument identification, description, options, Accessories
, specifications, and other basic information.
1-3. This manual is divided into eight sections which provide information as follows:
a. SECTION I, GENERAL INFORMATION, contains the instrument description and specifications as well as the accessory and recommended test equipment list.
b. SECTION II, INSTALLATION, contains information relative to receiving inspection, preparation for use, mounting, packing, and shipping.
c. SECTION III, OPERATION, contains operating instructions for the instrument,
d. SECTION IV, PERFORMANCE TESTS, contains information required to verify that instrument performance is in accordance with published specifications.
e. SECTION V, ADJUSTMENTS, contains information required to properly adjust and align the instrument after repair.
f. SECTION VI, REPLACEABLE PARTS, contains information required to order all parts and assemblies.
g. SECTION VII, MANUAL CHANGES, contains backdating information to make this manual compatible with earlier equipment configurations.
h. SECTION VIII, SERVICE, contains descriptions of the circuits, schematic diagrams, parts location diagrams, and troubleshooting procedures to aid the user in maintaining the instrument.
1-4. Supplied with this manual is an Operating Information Supplement. The Supplement is a copy of the first three sections of this manual, and should be kept
with the instrument for use by the operator. Additional copies of the Operating Information Supplement can be ordered through your nearest Hewlett-Packard office. The part number is listed on the title page.