|Dranetz Engineering Laboratories Inc. 305-PA-3002 Level Meter|
Dranetz Engineering Laboratories Inc.
model 305 Phasemeter (figure 1-1) represents a totally new concept in the measurement of phase angle. A precision phase angle measurement is obtained simply by connecting the signals and reading the display. No adjustments are required during normal use. model 305 senses the signal frequency and adjusts its output time constant for maximum speed commensurate with low ripple. It remains operational in the presence of up to 10% high frequency noise. Model 305 sensitivity to waveform distortion is reduced over that of other phasemeters which normally require filters to remove distortion effects. 1-5. The use of the automatic dual mode switching allows for continuous phase plotting between <-160° to > +340° with no discontinuities. The front panel controls on model 305 main frame are used only for self checking of the main frame and calibration of remotely connected Accessories such as a recorder or Printer. 1-6 . Standard Plug-In units covering a frequency range of 1 Hz to 11 MHz are available with model 305 Phasemeter. Special Plug-In units are available on request. 1-7. SPECIAL FEATURES 1-8. model 305 provides special features when used with Plug-In Model 305-PA-3001 or 305-PA-3002. a. No operating controls. b. Excellent stability -~ no operating adjustments. c. 0.05° accuracy independent of input voltage. d. Versatile Plug-In concept. e. Basic frequency range: 2 Hz to 700 kHz (extendable to I Hz or 11 MHz with other plug-ins). f. Digital readouts on Models 305B, 305C and 305D. g. Digital outputs on Models 305B, 305C and 3&5D-for use with printers, etc. h. 1 Megohm input impedance. i. Levels extendable to 40 KV using Tektronix probes. j. Indicator lights on each input to indicate either low or high input levels (305-PA-3001 only). k. High overvoltage capability.
Deze handboeken zijn beschikbaar voor de bovengenoemde apparatuur:
|Dranetz Engineering Laboratories Inc. -- 305-PA-3002 -- Service-en gebruikershandleiding|
|Handleidingen Type:||Service-en gebruikershandleiding|
|Maat:||6.32 Mbytes (6629316 Bytes)|
|Kwaliteit:||Gescande document, lezen deels slecht, deels niet leesbaar.|
|Upload datum:||11 augustus 2017|
|Downloads:||623 Sinds 11 augustus 2017|
X GENERAL INFORMATION 1-1 1-1 Introduction 1-1 1-3 Purpose 1-1 1-7 Special Features 1-1 1-9 Equipment Characteristics and Series 305 Configuration 1-2 II OPERATION 2-1 2-1 General 2-1 2-3 Controls, Connectors and Indicators 2-1 2-5 Unpacking 2-1 2-7 Operation 2-1 2-10 Main Frame Rear Connectors Terminal Designations 2-2 2-12 Operation of model 305 With Option 107 2-11 2-15 Additional Operating Features of Model 305D 2-12 2-17 Operation 2-12 III PERFORMANCE CHECK 3-1 3-1 General 3-1 3-3 Test Equipment Required 3-1 3-5 Main Frame Calibration 3-1 3-6 Minimum Performance Check 3-4 3-11 Field Installation of 305 Series Plug-Ins 3-7 IY THEORY OF OPERATION 4-1 4-1 General 4-1 4-3 Basic Operation of 360 Zero Crossing Phase Meters' 4-1 4-9 Block Diagram Discussion 4-2 Y MAINTENANCE 5-1 5-1 General 5-1 5-6 Mechanical Disassembly Procedure 5-2 5-8 Circuit Description 5-5 5-10 Model 305-PA-3001 Wide Range Plug-In 5-5 5-36 model 305 Main Frame Control Circuits 5-9 5-49 Model 305D Offset and Null Circuits 5-11 5-54 Series 305 Dual Power Supply 5-11 5-61 Selected Components 5-13 6-1 General 6-4 Equipment Required 6-5 Equipment Setup 6-6 Test Procedure 6-9 Adjustment Procedure for Option 107 6-11 (Card All) Adjustment Procedure for 305D Offset 6-13 Circuits Adjustment Procedure (Model 305-PA-3001 6-20 In Main Frame) Adjustment Procedure (Model 305~PA-3002 In Main Frame) REPLACEABLE PARTS LIST 7-1 General AVAILABLE OPTIONS 8-1 General 8-3 Option 101 8-6 Option 102 8-10 Remote Operation 8-18 Additional Operating Control Infor- 8-21 mation for Remote Operation of Automatic Calibration When Option 107 is Used Option 103 8-28 Option 105 8-31 Option 107
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.