Hewlett Packard 5328A Frequentieteller
Image 1 of 1
If you have any other photos or manuals for the
Hewlett Packard 5328A you can upload the files here .
The standard 5328A mainframe measures frequency, period, period average, time interval, time interval average, and ratio. A totalize function is also provided. The capabilities of the instrument are expanded (as shown in Table 1-3) by installing the options described in paragraph 1-17. The 5328A provides an 8-digit (9 with Option 030) LED display, display storage, and leading zero blanking. Decimal point and unit readouts are displayed automatically. Two independent selectable input channels are provided for time interval measurements. Each input channel has an attenuator, trigger slope selector, level control, ac-dc coupling, and an Oscilloscope marker output. Rear panel connectors provide a gate output, a time base output and an input for an external Frequency Standard. An ARM switch on the rear panel allows arming by the signal being measured (switch OFF) or by another input signal (switch ON).
18.33 Mbytes (19222521 Bytes)
Gescand document alle leesbaar.
31 May 2015
This manual applies to Hewlett-Packard Model 5328A Universal Counters with serial prefix number Ί604Α. This manual provides service information for the Hewlett-Packard Model 5328A Universal Counter (including Option 010). A separate Installation and Service Manual is provided for each option supplied with the 5328A, Operating information for the 5328A (with options) is covered in the 5328A Users Manual. 1-3. This manual is divided into five sections containing the following information: SECTION I GENERAL INFORMATION covers a description of the counter, options, equipment supplied, Accessories available, and specifications. SECTION II THEORY OF OPERATION covers a description of the general operating principles of the Counter in reference to block and schematic diagrams of each assembly. SECTION III MAINTENANCE contains maintenance and service information including a list of assemblies, recommended test equipment, performance checks and adjustments. Troubleshooting procedures and flowcharts are included in this section. SECTION IV REPLACEABLE PARTS provides a complete list of replaceable parts and information for ordering parts. SECTION V MANUAL CHANGES AND OPTIONS contains information on manual changes and available options. Field installation of Option 010 Oven Oscillator is contained in this section. SECTION VI SCHEMATIC DIAGRAMS contains schematic diagrams and component locators. A description of reference designators, signal mnemonics and printed-circuit card identification is included.
2.47 Mbytes (2585274 Bytes)
Gescand document alle leesbaar.
09 January 2013
(Including Options 010, 011, 020, 021, 030, and 040)
Terms and Conditions
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.