Tektronix TVS600A Series Analyzer
Nessuna immagine disponibile!
Forse puoi caricare una foto per la
Tektronix TVS600A Series?
Se avete altri manuali per la
Tektronix TVS600A Series potete caricare i file qui .
The TVS600 Series
Waveform Analyzers are a family of C-size, double-wide
VXI modules suitable for use in a variety of test and measurement applications and systems. Many key features are listed below:
Four standard configurations with full-featured, 1 MW/50 W inputs: TVS641
have four channels, TVS621
have two channels.
A maximum realtime digitizing rate up to 5 GSample/second with an analog
bandwidth up to 1 GHz.
A maximum record length of 30,000 samples with 8-bit vertical resolution.
Full programmability using a SCPI command set.
Acquisition modes such as sample, envelope, and average.
A full complement of internal triggering modes such as edge and pulse, plus
VXI backplane and external trigger sources.
Tipo di manuale:
Manuale di servizio
6.64 Mbytes (6958512 Bytes)
ID del manuale:
Documento elettronico, senza ricerca, molto ben leggibile.
2019 09 22
This manual is divided into chapters, such as Specifications and Performance Verification. Within each chapter, the information is further divided into subsections such as Product Description and Equipment Required.
Subsections that contain procedures also contain introductions before the procedures. Be sure to read these introductions because they provide information that is needed to do the service correctly and efficiently. The following list provides a brief description of the chapters contained in this manual:
■ Specifications describes the waveform Analyzer
and the characteristics that apply to it.
■ Operating Information includes general information and operating instructions at the level needed to safely power on and service the waveform Analyzer
■ Theory of Operation contains circuit descriptions that support general service to the module level.
■ Performance Verification contains an automatic Software
procedure to verify that the waveform Analyzer
functions properly and meets warranted specifications.
■ Adjustment Procedures refers to the Performance Verification section for adjusting the waveform Analyzer
to meet warranted specifications.
■ Maintenance contains information and procedures for performing preventive and corrective maintenance. These instructions include cleaning, module replacement, and fault isolation to the module level.
■ Options describes standard Accessories
, optional Accessories
, and options that may be installed in the waveform Analyzer
■ Electrical Parts List refers you to Mechanical Parts List, where both electrical and mechanical modules are listed. See below.
■ Diagrams contains a block diagram and an interconnection diagram that are useful when isolating failed modules.
■ Mechanical Parts List contains an illustration that shows all replaceable modules, and a table that lists the replaceable modules, their descriptions, and their Tektronix