|Kikusui Electronics Corporation PLZ-150W Altro|
Kikusui Electronics Corporation
|Categoria:||Strumenti di misura|
Series PLZ-W Electronic Loads can be operated in constant resistance mode, constant current mode and switching action electronically, replacing the conventional sliding resistors (rheostats). Major features of the PLZ-W series are as follows: 1. Digital Meters. A digital voltmeter and an ammeter with an auto range of 3 1/2 digits. 2. A 1O-Turn Multi-Dial. Current values and resistance values may be set very precisely with the 10 turn multi-dial having a lock mechanism. 3. Function of Presetting. Current and resistance values may be preset. A. Built-in Oscillator for Switching Action. To be used in the constant current mode. Switching action occurs within the frequency range of 10 Hz - 1 kHz. 5. Display of Loading Power. Supplied power value is displayed in a digital figure and in the form of luminescent bar graph. 6. Remote Control Functions. When operated in the constant current mode the instrument may be remotely controlled by either an external voltage or an external resistance. The instrument may also be remotely controlled by an external resistance when operated in the constant resistance mode. 7. One-controlled Parallel Operation. It is possible to increase current capabilities by operating a number of units connected in parallel simultaneously under the control of a master unit. 8. Protection Circuits Include: Overvoltage, overcurrent, overpower, overheating, and input terminal connection in reverse polarity. The PLZ series is offered as a portable desk-top unit and may be integrated into a 19 inch or 500 mm Kikusui standard rack, with the use of the option brackets» 1-2. Specifications Power Requirements. Line voltage: 120 V AC i 10% 50/60 Hz single phase Power consumption: Approx. 35 VA (at 120 V AC) Input. Loading voltage: 4 - 60 V DC. Loading current: 0 - 30 A Maximum loading power: 150 W Operating ambient temperature range: Ö - 40°C (32 - 104°F) Operating ambient humidity range: 10% - 90% RH Cooling method: Forced air cooling by means of fan Isolation from ground: ±250 V DC Modes. Constant current mode: 0 - 30 A (1st range) 0 - 3 A (2nd range) .Constant resistance mode: 10 - 100 τηΩ (1st range) (minimum 0.12 Ω) 0.1 - 1 Ω (2nd range) 1 - 10 Ω (3rd range) 10 - 100 Ω (4th range)
Questi manuali sono disponibili per l'attrezzatura di cui sopra:
|Kikusui Electronics Corporation -- PLZ-150W -- Servizio e manuale utente|
|Tipo manuale:||Servizio e manuale utente|
|Dimensioni:||1.52 Mbytes (1597429 Bytes)|
|Qualità:||Documento scansionato, la lettura in parte male, non in parte leggibile.|
|Data di caricamento:||14 agosto 2015|
|Download:||209 dal 14 agosto 2015|
SECTION 1. GENERAL ... 1 1-1. Introduction ... 1 1-2. Specifications ... 2 SECTION 2. OPERATION ... 5 2-1. Notes Before Operation ... 5 2-2. Front and Rear Panel Descriptions ... 10 2-3. Initial Setting ... 13 2-4. Operation in the Constant Resistance Mode ... 14 2-5. Use of nxl0" Range in the Constant Resistance Mode ... 17 2-6. Graph of Resistance Range and Power Limit Relation ... 18 2-7. Operation in the Constant Current Mode ... 20 2-8. Operation in the Switching Mode ... 22 2-9. Operation in the Constant Voltage Mode ... 24 2-10. Indication of Loading Power ... .. 27 2-11. Presetting the Power Lindt ... 28 2-12. Remote Control with an External Resistance (Constant Resistance or Constant Current) ... 29 2-13. Remote Control with an External Voltage (Constant Current only) ... 30 2-14. Remote Control of Load Switch by an External Terminal ... 31 2-15. One-control Parallel Operation ... 31 SECTION 3. OPERATING PRINCIPLES ... 34 3-1. Constant Current Operation ... 34 (j> 3-2. Constant Resistance Operation ... 34 ^ 3-3. Block Diagram ... 36 SECTION 4. MAINTENANCE ... 37 4-1. How to Open the Chassis Case ... . 37 4-2. Inspections and Cleaning ... 37 4-3. Calibration ... 38
These terms and conditions contain rules about posting comments. By submitting a comment, you are declaring that you agree with these rules:
- Although the administrator will attempt to moderate comments, it is impossible for every comment to have been moderated at any given time.
- You acknowledge that all comments express the views and opinions of the original author and not those of the administrator.
- You agree not to post any material which is knowingly false, obscene, hateful, threatening, harassing or invasive of a person's privacy.
- The administrator has the right to edit, move or remove any comment for any reason and without notice.
Failure to comply with these rules may result in being banned from further commenting.
These terms and conditions are subject to change at any time and without notice.