Fluke 8860A Multimètre numérique
5-1/2 digit Digital Multimeter
is a fully-guarded, microprocessor-controlled,
5-1/2 digit, five-function Digital multimeter
bench-top applications. Its measurement functions include:
VDC, true RMS VAC (ac coupled), true RMS VAC (decoupled),
two-terminal ohms, and four-terminal ohms. Each voltage
function is .provided with five ranges. Six ranges are
available for each of the ohms functions. The ranges can be
selected automatically (autorange) or manually.
Several other performance features are provided with the
8860A. They include:
• Total push-button control
• Dual slope a/d conversion
• Push-button zero for V dc and ohms
• Selectable sample rate and resolution (4-1/2 or 5-1/2 digits)
• Selectable internal/external guard connection
• Selectable trigger mode: auto, manual, external
• Three math functions: offset, limits and peak-to-peak.
• Numeric data entry push-buttons for use in loading the
constants for the offset, limits, and peak-to-peak functions.
• A 5-1/2 digit measurement display (mantissa) field with
sign and floating decimal point.
• A two-digit exponent field with sign for displaying
exponents of 10 and high, low, pass (H,L,P) limits
information. Exponents are automatically displayed for
• Measurement units annunciators that operate in conjunction
with the display decimal point to display the selected range.
• Overload protection for all functions and ranges.
• Selectable local/remote operation.
is housed in a plastic Portable Test Instrument
(PT1) case. The case includes a retractable carrying handle
and a fold-down bail. When extended, the bail provides a
convenient tilt for bench-top operation. The PT1 design
allows the 8860A
to be stacked and latched with other
instruments in the Fluke
PT1 product line. Stacking saves
bench space and allows the instruments in the stack to be
moved as a single unit.
Operating power for the 8860A
is selectable for the
following configurations: 100, 120,220, or 240V ac ±10%; 50
or 60 Hz. The unit is shipped in the configuration specified
at the time of purchase. Since the voltages are selectable,
they can be reset in the field by a qualified technician.
(Instructions for line voltage selection are given in the
Manuel de l'utilisateur
1.76 Mbytes (1847919 Bytes)
01 avril 1980
Document scanné, la lecture en partie mal, en partie illisible.
Date de transfert:
15 janvier 2018
The Operator Manual is divided into the following sections:
1 INTRODUCTION AND SPECIFICATIONS Introduces the 8860A
Instruction Manual Set, describes the 8860A
features, and lists the instrüment specifications.
2 SHIPPING AND SERVICE INFORMATION Provides unpacking and
information. Instructions for contacting your local Fluke
Service Center and the John Fluke
Mfg. Co., Inc. are also
3 INSTALLATION AND MAINTENANCE Bench-top and rack-panel
mounting procedures are followed by an input line voltage
verification procedure and operator maintenance procedures.
4 OPERATING INSTRUCTIONS Includes an illustrated
description of the 8860A
controls and indicators, detailed
operating notes, and operating procedures.
5 OPTION AND ACCESSORY INFORMATION Provides a description
and the specifications for each of the options and
available for use with the 8860A
instructions are also provided for the options. Accessory
operating information is provided with the individual