|Zeiss Superlux 300|
|Category||Medical and Biomedical Equipment|
|Description||High-Intensity Light Source|
General The Superlux 300 high-intensity light source consists of the lamp house (1) and the power supply unit (2). The light travels from the lamp house to the microscope or endoscope over a light guide. The xenon lamp of the Superlux 300 generates light whose spectrum is near that of natural daylight. The color temperature of the light remains constant regardless of the brightness setting. You may thus use daylight film in photographic documentation without additional conversion filters. The brightness can be adjusted at the lamp house. The xenon lamp can be made to "flash” to a higher intensity when making a photographic exposure. The lamp house also contains a 12 V, 100 W halogen lamp as a backup illumination system. If the xenon lamp ever goes out, you can continue surgery without delay. All that you have to do is change the connection of the light-guide and switch the selector (1.2) on the lamp house to "HALOGEN". The electrical supply for the halogen lamp is provided on the microscope stand or from an external power supply unit. The on/off switch and the brightness adjustment element for the halogen lamp is located on the microscope stand or on the external power supply unit.
These manuals are available for the above equipment:
|Zeiss -- Superlux 300 -- User Manual|
|Manual Type||User Manual|
|Size||3.49 Mbytes (3664209 Bytes)|
|Date||16 December 1993|
|Quality||Scanned document, all readable.|
|Upload date||20 February 2018|
|Downloads||96 since 20 February 2018|
Notes on safety ... 4 Notes on installation and usage ... 4 Warning signs and notes ... 6 General ... 7 Control elements, connections and displays ... 9 Lamp house ... 9 Power supply unit ... 11 Preparations for use ... 12 Operation of the instrument ... 13 Xenon illumination ... 13 Halogen illumination ... 13 Triggering the camera ... 14 Pre-surgery checklist ... 14 Using the instrument during an operation ... 15 Xenon lamp backup procedure ... 15 Maintenance ... 16 Exchanging the xenon lamp module ... 16 Exchanging the halogen lamp ... 17 Setting the instrument to the available line voltage / Changing fuses ... 17 Troubleshooting table ... 18 Installation possibilities on stands / Retrofit kits and parts .. 20 Retrofit kits for stands / mounts ... 20 Control units for photography ... 21 Spare parts ... 21 Specifications ... 22 Important for safety Page Notes on safety ... 4 Pre-surgery checklist ... 14 Troubleshooting table ... 18
No comments yet. Be the first!