|Kepco ÄPH 500M Power supply|
Kepco Series APH Power Supplies are high voltage, low current, precision-stabilized d-c sources of approximately twenty watts of d-c output power. Their important design features include; a) AUTOMATIC CROSSOVER BETWEEN VOLTAGE AND CURRENT STABILIZATION MODE OF OPERATION WITH VISUAL FRONT PANEL (LED) MODE INDICATORS, b) PRECISION, TEN-TURN FRONT PANEL OUTPUT VOLTAGE AND OUTPUT CURRENT CONTROLS. c) RECESSED FRONT PANEL OUTPUT METERS PERMIT THE SIMULTANEOUS MONITORING OF OUTPUT VOLTAGE AND OUTPUT CURRENT. d) FRONT AND REAR OUTPUT TERMINALS WITH REMOTE SENSING PROVISION, ALLOW PRECISION STABILIZATION AT A DISTANT LOAD. e) PROGRAMMING TERMINALS PERMIT ACCESS TO A BUILT-IN, INDEPENDENT PREAMPLIFIER, TO THE VOLTAGE CONTROL AND MAIN VOLTAGE AMPLIFIERS AND TO A SEPARATE CURRENT CONTROL AMPLIFIER. A MODE FLAG SIGNAL OUTPUT (TTL COMPATIBLE), AS WELL AS TWO D-C REFERENCE SOURCES (±6.2 volts at 1 mA MAX.} REFERENCED TO COMMON (+ SENSE) ARE PROVIDED. 1-5 The APH Power supply can, in simplified form, be electrically represented by the fold-out diagram in Section III of this manual (see FiG. 3-15 at the end of Section III). A high voltage, low current d-c source is controlled by a series pass element (vacuum tube), which in turn is controlled by the driver circuit via the voltage channel or the current channel. The driver is connected to three control amplifiers (Main Voltage, Internal Current Control, External Current Control) via the gating circuit, which permits only one Amplifier to take control of the APH output.
These manuals are available for the above equipment:
|Kepco -- ÄPH 500M -- Service and User Manual|
|Manual Type||Service and User Manual|
|Size||2.80 Mbytes (2940958 Bytes)|
|Quality||Scanned document, all readable.|
|Upload date||14 August 2015|
|Downloads||185 since 14 August 2015|
This manual contains instructions for the installation, operation and maintenance of the APH High Voltage, Automatic Crossover Power Supplies.
|Start Free Download|
No comments yet. Be the first!