SERIAL PREFIX: 1616A
This section applies directly to Model 5305B 1300 MHz
Counters having Serial Prefix 1616A. This section is
provided in loose-leaf form for incorporation into the 5300
Measurement System Manual. 5305A instruments are
documented in a separate manual.
This section with enclosed "Manual Changes" sheet
applies directly to HP Model 5305B 1300 MHz Counters
having Serial Prefix numbers above 1616A.
Subsection VII of this document contains information
pertinent to all older instruments.
This manual provides operating and service
information for the Hewlett-Packard Model 5305B 1300
MHz Counter. Information for the mainframes (5300A
or 5300B) is contained in separate manuals. This
manual is divided into eight sections containing the
SECTION 1 GENERAL INFORMATION covers a
description of the counter, equipment supplied,
accessories, specifications, and recommended test
SECTION II INSTALLATION provides instructions
for unpacking, inspection, preparation for use,
shipment, and storage for the counter.
SECTION III OPERATION covers the counter's
operating features including front-panel controls,
input level considerations, and operating and selfcheck
SECTION IV THEORY OF OPERATION
describes the counter's theory of operation.
SECTION V MAINTENANCE contains an incabinet
performance check, adjustments, and
SECTION VI REPLACEABLE PARTS provides a
complete list of the counter's replaceable parts and
information for ordering parts.
SECTION VII MANUAL CHANGES provides
information necessary to backdate the manual to cover
SECTION VIII SCHEMATIC DIAGRAMS THEORY
contains schematic diagrams, and component locators.