|Hewlett Packard 5305B Counter|
|Description||1300 MHz Counter|
The 5305B extends the frequency measuring capability of the 5300 Measuring System to the UHF range. The counter features burst or CW measurements to 1300 MHz, separate channels to cover 90 MHz-1300 MHz and 50 Hz to 100 MHz both with 20 mV rms sensitivity, high resolution mode for fast tone measurements, automatic gain control for both channels or manual attenuator control for the high frequency channel, fuse protected high frequency channel, and probe power plus accessory preamp for high sensitivity applications. When operating in the high resolution mode, a phaselocked multiplier gives 1000 times improvement in the resolution of audio tone measurements. This feature is especially useful for servicing equipment using tone modulation for digital encoding on the carrier.
These manuals are available for the above equipment:
|Hewlett Packard -- 5305B -- Service and User Manual|
|Manual Type||Service and User Manual|
|Size||40.45 Mbytes (42414560 Bytes)|
|Quality||Scanned document, all readable.|
|Upload date||08 December 2013|
|Downloads||259 since 08 December 2013|
SERIAL PREFIX: 1616A This section applies directly to Model 5305B 1300 MHz Counters having Serial Prefix 1616A. This section is provided in loose-leaf form for incorporation into the 5300 Measurement System Manual. 5305A instruments are documented in a separate manual. NEWER INSTRUMENTS This section with enclosed "Manual Changes" sheet applies directly to HP Model 5305B 1300 MHz Counters having Serial Prefix numbers above 1616A. OLDER INSTRUMENTS Subsection VII of this document contains information pertinent to all older instruments. This manual provides operating and service information for the Hewlett-Packard Model 5305B 1300 MHz Counter. Information for the mainframes (5300A or 5300B) is contained in separate manuals. This manual is divided into eight sections containing the following information: SECTION 1 GENERAL INFORMATION covers a description of the counter, equipment supplied, accessories, specifications, and recommended test equipment. SECTION II INSTALLATION provides instructions for unpacking, inspection, preparation for use, shipment, and storage for the counter. SECTION III OPERATION covers the counter's operating features including front-panel controls, input level considerations, and operating and selfcheck procedures. SECTION IV THEORY OF OPERATION describes the counter's theory of operation. SECTION V MAINTENANCE contains an incabinet performance check, adjustments, and troubleshooting information. SECTION VI REPLACEABLE PARTS provides a complete list of the counter's replaceable parts and information for ordering parts. SECTION VII MANUAL CHANGES provides information necessary to backdate the manual to cover earlier instruments. SECTION VIII SCHEMATIC DIAGRAMS THEORY contains schematic diagrams, and component locators.
|Start Free Download|
No comments yet. Be the first!