|Hewlett Packard 8554B|
|Description:||Spectrum Analyzer RF Section|
The HP Model 8554B Spectrum analyzer RF Section covers the frequency range from 100 kHz to 1250 MHz. When it is combined with an IF Section and a Display Section it functions as the tuning section of an RF spectrum analyzer. The analyzer electronically scans input signals and displays their frequency and amplitude on a CRT. The horizontal, x-axis, is calibrated in units of frequency and the vertical, y-axis, is calibrated in absolute units of voltage (/j.V or mV) or power (dBm). Absolute or relative measurements of both amplitude and frequency can be easily made.
These manuals are available for the above equipment:
|Hewlett Packard -- 8554B -- Service and User Manual|
|Manual Type:||Service and User Manual|
|Size:||23.35 Mbytes (24483679 Bytes)|
|Quality:||Scanned document, reading partly badly, partly not readable.|
|Upload date:||27 November 2013|
|Downloads:||136 since 27 November 2013|
This manual applies directly to instruments with serial numbers prefixed 1245A. For additional important information about serial numbers, see INSTRUMENTS COVERED BY MANUAL in Section I. Section II, Installation: provides information about initial inspection, preparation for use, and storage and shipment. Section III, Operation: provides information about panel features, and provides operating checks, instructions, and maintenance information. Section IV, Performance Tests: provides the information required to verify that the instrument is performing as specified in Table 1-1. Section V, Adjustments: provides the information required to properly' adjust and align the instrument. Section VI, Replaceable Parts: provides ordering information for all replaceable parts and assemblies. Section VII, Manual Changes: this section is reserved to provide manual change information in future revisions of this manual. Section VIII, Service: provides the information required to repair the instrument.
|Start Free Download|