|Tektronix 465M Oscilloscope|
These manuals are available for the above equipment:
|Tektronix -- 465M -- Service and User Manual|
|Manual Type||Service and User Manual|
|Size||45.30 Mbytes (47498145 Bytes)|
|Date||15 March 1977|
|Quality||Scanned document, all readable.|
|Upload date||26 April 2013|
|Downloads||892 since 26 April 2013|
1-1. INTRODUCTION a. Manual Purpose. This manual provides instructions for operation and maintenance of the AN/USM-425(V)1 Oscilloscope and includes illustrated parts breakdown data. The AN/USM-425(V)1 is a Model 465M manufactured by Tektronix, Incorporated, Beaverton, Oregon. b. Manual Scope. The instructions provided in this manual are intended to be performed at organizational or intermediate level maintenance activities using tools, test equipment, and spare parts authorized in their allowance lists and supply activities. c. Manual Arrangement. This manual is separated into eleven sections as follows: (1) Section I, Introduction and General Information. Contains the purpose, scope, and arrangement of the manual and a description of the instrument including its leading particulars and accessories. (2) Section II, Special Tools and Test Equipment. Contains listings of tools, test equipment, and consumable materials needed to maintain the instrument. (3) Section III, Preparation for Use and Shipment. Contains instructions for preparing the instrument for initial use and repackaging for shipment. (4) Section IV, Operation Instructions. Contains instrument theory of operation; a description of controls, connectors, and indicators; special operating instructions; initial turn-on and adjustment procedures; normal operation familiarization procedures; and examples of instrument applications. (5) Section V, Maintenance Instructions. Contains procedures to check out, perform routine maintenance, troubleshoot, repair, test, and adjust the instrument. (6) Section VI, Diagrams. Contains wiring and schematic diagrams with associated data referenced in other sections of the manual. Also, may contain some illustrations to improve operation and maintenance of the instrument. (7) Section VII, Introduction to the Illustrated Parts Breakdown. Contains information on how to use the illustrated parts breakdown data in Sections VIII through X. (8) Section VIII, Maintenance Parts List. Contains the illustrated parts breakdown illustrations and parts descriptions.
|Start Free Download|
No comments yet. Be the first!