|Advantest Corporation R3182 Spectrum analyzer|
The R 3132 Series spectrum analyzer uses the synthesized local method and allow the user highly stable and highly precise spectrum analysis. The key features of the R3132 Series spectrum analyzer are listed below. (1) Frequency Range: Frequency span: 9 kHz to 3 GHz (R3132/N) 9 kHz to 8 GHz (R 3162) 9 kHz to 26.5 GHz (R3172) 9 kHz to 40 GHz (R3182) Zero, 1 kHz to 3 GHz (R3132/N) Zero, Ì kHz to 8 GHz (R3162) Zero, 1 kHz to 26.5 GHz (R3Ì72) Zero, 1 kHz to 40 GHz (R3182) (2) High-speed and high-precision sweep Frequency span accuracy: 1% or less Sweep time: 20 ms (3) Wide dynamic range Maximum input level: +30 dBm Built-in pre-amplifier: -132 dBm noise level (4) Frequency counter function with a resolution of 1Hz. (5) A power measurement function useful for evaluating radio instruments using measurements such as occupied bandwidth (OBW), adjacent channel power (ACP), channel power, etc. (6) An auto tuning function that searches for a signal with the maximum input level. (7) Save and recall functions which you can use to store measurement conditions and data in TEXT for mat. (8) A 3.5-inch floppy disk drive which you can use to save screen images in BMP format. (9) Support for ESC/P, ESC/P-R and PCL compatible printers. (10) Remote control capabilities which allow you to setup an automatic measurement system. This remote control function complies with GPIB and RS-232 specifications. (11) High-precision color LCD
These manuals are available for the above equipment:
|Advantest Corporation -- R3182 -- User Manual|
|Manual Type||User Manual|
|Size||9.43 Mbytes (9888259 Bytes)|
|Date||20 March 1999|
|Quality||Scanned document, all readable.|
|Upload date||17 August 2016|
|Downloads||266 since 17 August 2016|
. INTRODUCTION • Product Description • Accessories • Options • Operating Environment • System Checkout • Cleaning, Storing and Transporting the R3132 Series Spectrum Analyzer • Calibration • Replacing Parts with Limited Life Includes a description of the spectrum analyzer and its' parts along with information on its’ operating environment and how to perform a system checkout. 2. OPERATION • Panel Description • Basic Operation • M easurement Examples • Other Functions Describes the names and the functions of each part on the panels. You can learn the basic operation of the spectrum analyzer through the examples shown in this chapter. 3. REFERENCE * Menu Index * Menu Map • Menu Function Descriptions • List of Settings Shows a list of operation keys, and describes the function of each key. 4. REMOTE PROGRAMMING • GPIB Command Index • GPIB Remote Programming • RS-232 Remote Control Function Gives an outline of the GPIB and RS-232 interfaces, and how to connect and set them up. Also included are a list of commands necessary for programming and using the program examples. 5. PERFORMANCE VERIFICATION • General • Procedures of Performance Verification • Tracking Generator Performance Verification Procedure • Performance Verification for OPT73 (FM Demodulation) • Performance Verification Record Sheet Describes the performance verifica tion. 6. PERFORMANCE • External • External • External • External Describes the performance verifica- tion(External Mixer). VERIFICATION (External Mixer) Mixer OPTI 6 Mixer OPT 17 Mixer OPT18 Mixer OPT19 7. SPECIFICATIONS • R 3 132 Specifications • R 3Ì32N Specifications * R 3 162 Specifications • R3172 Specifications * R 3 182 Specifications • Options • Shows the specifications o f the spectrum analyzer.
|Start Free Download|
No comments yet. Be the first!