|Knight-Kit LOW CAPACITY PROBE 38K153 Probe|
|Model||LOW CAPACITY PROBE 38K153|
|Description||LOW CAPACITY PROBE|
The input system of an oscilloscope contains a certain amount of resistance shunted by capacitance. This is called the input impedance of the oscilloscope. To the input impedance of the oscilloscope is also added the capacity of the test leads. This total shunting capacitance and resistance, when connected across a high impedance, high frequency, or broad band width circuit, may be sufficient to detune the resonant circuit or to load the test circuit and seriously distort the waveforms under observation. For this reason a low capacity (high impedance) probe is used to isolate the input of the oscilloscope from the circuit under test when waveforms are to be observed in any audio, video, sweep or sync circuit.
These manuals are available for the above equipment:
|Knight-Kit -- LOW CAPACITY PROBE 38K153 -- User Manual|
|Manual Type||User Manual|
|Size||359.56 Kbytes (368188 Bytes)|
|Quality||Scanned document, all readable.|
|Upload date||15 August 2015|
|Downloads||91 since 15 August 2015|
|Start Free Download|
No comments yet. Be the first!