SERIAL PREFIX: 2112
This Operating and Service Manual contains the information
required to operate, test, adjust, and service the
Hewlett-Packard Model 5005A Signature Multimeter. The
multimeter with its supplied accessories is shown in Figure 1-1.
1-3. MANUAL SUMMARY
1-4. This manual is divided into eight sections, each
covering a particular topic for the operation and service of
the HP 5005A. The topics by section number are:
SECTION I, GENERAL INFORMATION. Provides the instrument
specifications, instrument identification, accessories and
recommended test equipment.
SECTION II, INSTALLATION. Provides information about initial
inspection, preparation for use, storage and shipment.
SECTION III, OPERATION. Provides information about operating
characteristics, panel features, Operator’s Check, operating
instructions, measurement procedures, and maintenance.
SECTION IV, PERFORMANCE TESTS. Provides abbreviated
procedures for operational verification which give the
operator a high degree of confidence that the 5005A is
operating properly. Also provides expansive performance test
procedures which test the electrical performance of the
5005A, using the specifications in Table 1-1 as standards.
SECTION V, ADJUSTMENTS. Provides the procedures and
adjustment locations required to properly maintain the
instrument operating characteristics within specifications.
SECTION VI, REPLACEABLE PARTS. Provides ordering information
for all replaceable parts and assemblies within the instrument.
SECTION VII, MANUAL CHANGES. This section is reserved for
manual change information which effectively "backdates” the
technical areas of the manual to apply to older instruments.
SECTION VIII, SERVICE. This section provides the instrument
theory of operation, troubleshooting information, repair
techniques, and schematic diagrams.