|Hewlett Packard 5005A Digital multimeter|
The 5005A Signature Multimeter is a multipurpose instrument for troubleshooting electronic logic circuits to the component level. The 5005A can display digital “signatures” of logic circuits. This method of troubleshooting is called “signature analysis”. Typically a logic product designed for signature analysis troubleshooting will have a programmed controller and a stored or externally-provided test program which can exercise most of the unit. 1-9. The 5005A also measures frequency, pulse counts, time intervals, DC voltages, voltage differences, positive or negative peak voltages, and resistances.
These manuals are available for the above equipment:
|Hewlett Packard -- 5005A -- Service and User Manual|
|Manual Type||Service and User Manual|
|Size||20.35 Mbytes (21342179 Bytes)|
|Quality||Scanned document, all readable.|
|Upload date||30 May 2015|
|Downloads||232 since 30 May 2015|
SERIAL PREFIX: 2112 This Operating and Service Manual contains the information required to operate, test, adjust, and service the Hewlett-Packard Model 5005A Signature Multimeter. The multimeter with its supplied accessories is shown in Figure 1-1. 1-3. MANUAL SUMMARY 1-4. This manual is divided into eight sections, each covering a particular topic for the operation and service of the HP 5005A. The topics by section number are: SECTION I, GENERAL INFORMATION. Provides the instrument specifications, instrument identification, accessories and recommended test equipment. SECTION II, INSTALLATION. Provides information about initial inspection, preparation for use, storage and shipment. SECTION III, OPERATION. Provides information about operating characteristics, panel features, Operator’s Check, operating instructions, measurement procedures, and maintenance. SECTION IV, PERFORMANCE TESTS. Provides abbreviated procedures for operational verification which give the operator a high degree of confidence that the 5005A is operating properly. Also provides expansive performance test procedures which test the electrical performance of the 5005A, using the specifications in Table 1-1 as standards. SECTION V, ADJUSTMENTS. Provides the procedures and adjustment locations required to properly maintain the instrument operating characteristics within specifications. SECTION VI, REPLACEABLE PARTS. Provides ordering information for all replaceable parts and assemblies within the instrument. SECTION VII, MANUAL CHANGES. This section is reserved for manual change information which effectively "backdates” the technical areas of the manual to apply to older instruments. SECTION VIII, SERVICE. This section provides the instrument theory of operation, troubleshooting information, repair techniques, and schematic diagrams.
|Start Free Download|
No comments yet. Be the first!