|Philips PM3340 Oscilloscope|
|Description:||2 GHz Digitizing Oscilloscope|
This compact dual channel digitizing storage oscilloscope features sampling techniques that allow display of signals with a frequency up to 2 GHz. The sampling .system can work in normal and in high resolution eye pattern mode. The instrument is extremely easy to use because of the AUTO-SET pushbutton, that automatically adjusts the controls to suit the input signal value. The brightness is independent of the time base settings . The M68000 microprocessor gives a wide choise of measurement and display possibilities, which can be selected via the ergonomic designed front panel, Figure 2.1 2 GHz digizing oscilloscope. The oscilloscope is provided with integrated circuits (including thin- film circuits), which guarantee a highly stable operation. Furthermore, connection to the local mains is simplified by a tapless switched-mode power supply that covers most voltage ranges in use: 90 V. . .264 V a.c. All these features make this oscilloscope suitable for a wide range of high-frequency measuring applications such as in telecommunication, component testing and development and in fast computers.
These manuals are available for the above equipment:
|Philips -- PM3340 -- User Manual|
|Manual Type:||User Manual|
|Size:||8.47 Mbytes (8884544 Bytes)|
|Manual-ID/Number:||4822 872 00368|
|Date:||01 March 1988|
|Quality:||Scanned document, all readable.|
|Upload date:||18 May 2015|
|Downloads:||93 since 18 May 2015|
ARRANGEMENT OF MANUAL This Operating Manual is arranged in such a way that the essential information on safety and operating procedures is immediately available in the first chapters. You are strongly advised to read Section 3.2. SAFETY INSTRUCTIONS thoroughly before installing your oscilloscope. Operating information is given in the remainder of Chapter 4. Complete information for preventive maintenance on the instrument can be found in Chapter 5. This is followed by details of the functional, mechanical and environmental data, listed in Chapter 6, Characteristics. Finally, additional information regarding the various versions of the instrument and accessories is given in Chapter 7, and Chapter 8. respectively.
|Start Free Download|