Applicable Serial Nos. start from MT05177
This manual explains the troubleshooting and repair
procedures for the MS710C/D/E/F (called MS710[ ] hereafter)
Spectrum Analyzer. It contains the following contents:
Section 2 explains how to disassemble each part
necessary when checking a suspected faulty block.
This section also includes the parts list of mechanical
Section 3 explains the entire signal flow for each
part based on the MS710[ ] block diagram, and provides
the basic knowledge required for MS710[ ] troubleshooting.
Section 4 explains how to locate a faulty block according
to the failure symptoms. This section also contains the
overall circuit diagram of the MS710[ ] (interblock
Section 5 contains circuit diagrams and a detailed
description of each block (PC board). The voltages,
signal levels, and waveforms are explained at the
check-points for troubleshooting. The adjustment
procedure for each block is also included.
Section 6 explains the replaceable parts list and
precautions when ordering.